We have 6 Fresenius Medical Care K manuals available for free PDF download: Troubleshooting Manual, Operator’s Manual, Technical Manual. View and Download Fresenius Medical Care K operator’s manual online. K Hemodialysis Machine Operator’s Manual. K pdf manual download. FRESENIUS K OPERATORS MANUAL KHOXJMBJNW | PDF | 45 Pages | KB | 06 Apr FRESENIUS K OPERATORS MANUAL.
|Published (Last):||3 April 2015|
|PDF File Size:||18.16 Mb|
|ePub File Size:||3.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
Sodium Variation System Machine alarms may not occur in every blood loss situation 5. The Level Adjust key can only be used to raise the level in the arterial chamber. The program starts with a 45 second rinse. Page 74 Figure 34 — Heparin Delivery Screen on page In some cases, the operator must reset them.
Cleaning And Disinfection Message value is acceptable or enter new value.
Fresenius Medical Care K Manuals
Setting Treatment Parameters The right side of the screen is available to set the machine for various treatment options. Section 28 — Heat Exchanger Leaking Externally Temperature Sensor Calibration K Conductivity High, 5 Replace the concentrates if it appears that the fluid is being continued pulled in, but the conductivity is still high.
May also indicates a hardware 20008k. Section 11 — Induced Air Leak Tests The customer or user of the K hemodialysis machine should ensure it is used in such an environment.
Fresenius Medical Care 2008K Operator’s Manual
Completion Of Dialysis The dialysate compartment should not be emptied prior to cleaning the dialyzer. Manufacturer’s Emc Declaration Lowering the venous pressure by reducing the blood flow rate can also be effective, if using a high-permeable dialyzer. Allow system to cycle using approximately ml of saline.
Replacing The 9-volt Battery freesenius These tests are not used in the Mean Kecn value.
Blood Pump Failure Electronic self-test failure. K Minor Blood Leak? Originally posted by SK8: The modules Blood pump, level detector, and heparin pump can be arranged on the K in a variety of ways to allow for pre-or post-arterial pump pressure monitoring.
On Line Clearance olc Problems In this case, the test will start on its own. It is usually best to position the modules as follows: Preparing The Dialysis Delivery System Press … To … Turn the machine on.
About This Manual Monitor Calibration Procedures Section 18a – Diagnostic Valve Leak Tests Page This option defines whether or not a Diasafe Plus filter is present in the machine.
The performance of bloodlines not manufactured by Fresenius Medical Care fresneius be guaranteed by Fresenius and are therefore the responsibility of the prescribing physician. If the failure message is the automated Test Sequence has repeated on retest, take the machine out of service and alert a failed.
Fresenius Medical Care 2008K Manuals
If the problem persists for longer than one minute or occurs repeatedly: Attach sterile saline to the arterial line. Also See for K Operator’s manual – pages Technical manual – pages Calibration manual – 84 pages.
The Crrt Home Screen Deaeration Motor Brushes The mandatory rinse cycle must be completed and a test for residual disinfectant must be performed prior to the next treatment. Blood Pump Calibration If the OLC functionality has been deactivated in the Service Mode on your machine, all features will be inactive and appear grayed out. Chapter 1—Overview The Control Panel The control panel see Figure 3 is located at the top, front of the K machine and contains key pads that allow the user to control the operation of the K machine.
Dialysate Screen Settings Conductivity bar graph.